Ismael aviso electric feb 12 2011 part 2 of 2 phenomina validation
I am using analog calibrated 50 amp meter range. based from the idle test from starting the amps draw reach more than 150amps, but once the ambient start to comes in it’s go down to 2 – 3 amps. at idle & we should expect & estimate x 2 – x4 at running. my system Energy ambient collection with different level. Now I can estimate we are at 35% , but if I will increase the speed w/o the proper resonant tune up, the result completely reverse, since the resonant will work as additional to the load. Like what’s happening at starting. Resonant start to fully activate after 1 second to give time on scanning of available ambient freq. I must design at least a 3 milliseconds speed scanner , the same with the cellphone unit scanner. Cellphone freq. can scan less 5 ms. for 15 freq. still a lot to do.